The AmeriGlide Vesta
The Vesta by AmerGlide is an indoor, straight stairway lift. The lift is a battery-powered model and works off of two 12 Volt batteries that require a continuous charge. If the lift stops working there is also a digital display located on the footrest for basic problem-solving. For operating codes displayed on the diagnostic display, refer to the following table to help diagnose and correct any faults. This lift should only be serviced by a technician familiar with the Vesta brand and needes to be checked on an annual basis or every 6-months if used frequently.
List of operating codes for the Vesta chairlift.
If the recommended corrective action provided in the table does not solve the problem, contact your authorized dealer for assistance.
|Code||Status/Fault||Recommended Corrective Action|
|No display.||Press the UP/DOWN rocker switch and see if the display comes on. Check that the key switch and ON/OFF switch are turned on.|
|-||On when the stairlift is okay; ready to use.|
Flashing when the stairlift is running.
|No action required.|
|0||Final limit switch is activated or swivel|
seat switch is activated.
|Check that the key switch is ON and the seat is fully swiveled to the riding position.|
|1||Batteries require charging.||Check that the charger is plugged in and charging.|
|2||Batteries not charging.||Check that the charger is plugged in and charging.|
|3||Upper limit safety switch is activated.||You have reached the upper limit. Run the unit in the down direction.|
|4||Safety edge switch (bumper) in up direction is activated.||Remove any obstacle that may have activated the switch.|
|5||Underpan switch is activated.||Remove any obstacle from underneath the footrest.|
|6||Safety edge switch (bumper) in down|
direction is activated.
Lower limit safety switch is activated
|Remove any obstacle that may have activated the switch. You have reached the lower limit. Run the unit in the up direction.|
|7||Batteries approaching critical level and|
|Check that the charger is plugged in and charging.|
|9||Rocker switch is activated in UP/DOWN|
|No action required.|
|A||Hinge is open.||Call for service.|
|B||A switch is active at power on (toggle up,|
toggle down, IR up, IR down, or learn
|Call for service.|
|C||IR address (DIP switch) does not match||Call for service.|
|D||Main power relay is welded||Call for service.|
|E||Main power relay did not close.||Call for service.|
|G||Brake not connected.||Call for service.|
|H||Main power relay has closed early.||Call for service.|
|L||Current limit exceeded.||Call for service.|
|N||Stairlift is running at half speed.||Call for service.|
|O||Main control board has a memory problem.||Call for service.|
|P||Powered swivel did not respond.||Call for service.|
|R||No charging current.||Check that the charger is plugged in and charging.|